If you got here, it is because you have already entered test mode and want to check that the Teros 12 Sensors are connected correctly. If you do not know how to enter the test mode, click here.
at this moment your device will be showing some audible and luminous indications. It will remain in this mode until you restart it.
The light indications by LEDs and the sequence of ascending or descending sounds are the result of the test until that moment.
We have prepared a table so that you understand what each indication of the LEDs means. The meanings will be associated with each of the ports on satellite devices.
Satellite device ports
If your equipment does NOT have a Rain Gauge and the test result is only the green LED flashing it means that the soil sensors are working correctly.
However, if your device has a Rain Gauge, note that the third possible indication of results may mean that a test has not yet been performed.Then, when this combination appears, just perform the procedure described below and the testing of your equipment will continue.
Rain Gauge test:
Remove the rain collector cup
Move completely the tipping bucket ten times. (from one side the other)
If your equipment has no problem and the procedures are performed exactly as described the audible indication will become ascending.
However, if after the above procedure the result of your Rain Gauge test is showing the flashing lights and downward sound, remove the sensor connector and reinsert it in the device in the correct position( it must enter without forcing) and return to the step 1 above to start the test again.
If after performing all these steps and confirming that the connectors are properly connected and still error messages are displayed talk to your contact at Agrosmart or send us a message in the chat to report your problem.